Plethysmographs are used to measure lung volume and airway resistance via Boyle’s Law. The basic concepts are simple but there are numerous variations. Although considered by most to be a modern invention its principles go well back into history.
Animal Plethysmograph, 1891. From Physiologische graphik, By Oskar Langendorff, published by F. Deuticke , 1891, page 268. The plethysmograph is connected to a Marey tambour.
From Johns Scott Haldane and John Gillies Priestley. “The regulation of the lung‐ventilation.” The Journal of physiology 32.3-4 (1905): 225-266.
“In investigating the normal respiratory ventilation is is evidently important to avoid breathing through tubes, vales, etc,. and equally important to have a definitive quantitative record of the air breathed. In order to attain these two objects we employed a body plethysmograph arranged as follows (see Fig. 2).
“The box of the plethysmograph was of such a size and shape as to include as small a space as practicable, and yet enable the subject of the experiment to sit in perfect comfort. Its internal dimensions were as follows:–Floor, 45 inches long. Back, 19-1/2 inches wide x 24-1/4 inches. Lid, 19-1/2 inches wide behind x 16 inches in front x 14-3/4 inches. Sloping front, 37-1/4 inches x 16 inches x 10 inches. End, 10 inches wide by 12-1/2 inches high.”
“The lid was in two parts, with a circular aperature in the middle, through which the neck passed. Thumb-screws were provided for fastening it to the box, and all joints, screw-holes, etc., were made tight with “plasticene” (modelling wax). Air-tight connection was made between the neck of the subject and the lid of the box by means of a rubber collar with a broad flange, which was slipped over the head, and fitted the neck as tightly as possible without causing congestion. The flange was pressed down tightly on a layer of plasticene on the lid of the box by a ring of wood. This wooden ring was of course in two pieces, and was pressed down on the collar by means of thumb screws.
“In the side of the box were two holes into which tubes were fitted by means of corks. One of these was connected with a rubber tube for letting air into or out of the box to adjust the level of the writing-point and counteract, if necessary, the influence of rise of temperature in the box. The other was connected with the recording apparatus by means of rubber tubing 1 inch in diameter.
“The recording apparatus itself was arranged as shown in the figure. The cylincer suspended to the balance beam was of light tin plate. Two sizes were used — of 8 and 12 inches diameter — according to the depth of breathing expected. As the up and down movements of the writing-point traced on the smoked cylinder were proportional to those of the bell-jar.”
Plethysmograph, Boruttau, 1908. From: Lehrbuch der medizinischen Physik: Für Studierende und Ärzte zur Ergänzung jedes Lehrbuchs der Experimentalphysik by Heinrich Johannes Boruttau, published by Barth, 1908, page 131.
Plethysmograph, 1911. From Human Physiology, Volume 1, By Luigi Luciani, published by Macmillan and Company, 1911, page 422.
“The same effect may be obtained when the animal is breathing free air, while enclosed within a hermetically sealed glass cylinder (Knoll). A tube tied in the trachea, or fitting tightly over the mouth and nostrils of the animal, passes through the wall of one box and communicates with the external air. The internal air of the box is connected by means of a second tube with a recording tambour, and traces, like a plethysmograph, the variations in the total volume of the animal , corresponding to the inspiratory and expiratory movements. The simplest application of this method is that of Bernstein, represented in Fig. 188.
Infant Plethysmograph, 1931. From: J Clin Invest. 1931 August; 10(3): 545–558, BREATHING MEASUREMENTS ON NORMAL NEWBORN INFANTS, Douglas P. Murphy and Edward S. Thorpe, Jr.
Volume Displacement Plethysmograph, 1960. Built by Jere Mead and J.H. Emerson. From Mead J. Volume displacement body plethysmograph for respiratory measurements in human subject. J Appl Physiol 1960; 15:736-740. Article provided by Carl O’Donnell, PhD, MPH
Volume Displacement Plethysmograph, 1960. Found on the Francis A. Countway Library of Medicine website. Warren Anatomical Museum: 21595. Experimental human body plethysmograph manufactured by JH Emerson Co. Machine was used to measure lung capacity. A person would sit inside the box with his head in the glass dome on top and breathe into the tube on front of dome. Machine is wooden and “L” shaped, designed to fit around a seated human, and latches closed. A spirometer would have rested on wooden edge. Two thermometers affixed to front measured interior and exterior temperatures. Tubing, gauge, and motor are affixed to right side. Rear of machine has attached wires and area marked “Sears-port”. 83 D x 59 W x 152 H cm.
Plethysmograph, 1963. A constant pressure-variable volume plethysmograph that appears to have been based on the design of Mead and Emerson. From a doctoral thesis, “Acute effects of inhalation of cigarette smoke on pulmonary mechanics” by John M. Miller, University of Edinburgh, Edmonton, Canada, 1963, page 18.
Plethysmography, 1960’s. Found at Zeepreventorium.org. No description given. Probably from the 1960′s. It is likely a volume-displacement plethysmograph and may be made by Godart or Mijnhardt.
Plethysmograph, Canine, 1964. From “A Canine Inhalation Exposure Apparatus Utilizing a Whole Body Plethysmograph.” by Boecker, B. B.; Aguilar, F. L.; Mercer, T. T., Health Physics, Volume 10, Issue 12, December, 1964.
Med-Science Pulmo-Box Model 275, 1967. A volume-displacement plethysmograph. From Medical Electronic Laboratory Equipment 1967-68: Pergamon Electronics Data Series by G. W. A. Dummer, J. Mackenzie Robertson, published by Elsevier, page 826.
Plethysmograph, 1970’s. The manufacturer and exact date are unknown. Photo courtesy of James Sullivan, BA, RPFT, Supervisor, Pulmonary Laboratories, Memorial Sloan Kettering Cancer Center
Volume Displacement Spirometer, circa 1970’s. Possibly made by Godart or Minjhardt. Found on http://www.seslhd.health.nsw.gov.au/POWH/history
Collins Plethysmograph, 1976. From Medical Instrumentation for Healthcare, by Cromwell L, Arditti M, Weibell FJ, Pfeiffer EA, Steele B, Labok J. Published by Prentice-Hall, 1976. Page 266.
Collins Plethysmograph, circa 1980. Photo courtesy of Carl O’Donnell, PhD, MPH.
Fukuda Sangyo, PCF-4000, 1984: A transmural plethysmograph. The PCF-2000 was an optional digital computer for performing tests. Tests could be performed and calculated manually with just the PCF-4000.Photo from a sales brochure courtesy of James Sullivan, BA, RPFT, Supervisor, Pulmonary Laboratories, Memorial Sloan Kettering Cancer Center
Constant volume plethysmograph. Manufactured by Med Graphics in 1989. From a Pemed listing.
Collins Plethysmograph Box 1,1990. Photograph provided by Richard Johnston
Collins Infant Plethysmograph, circa 1995. Found at MedUsed.com.
Sensormedics Constant Volume Plethysmograph, 2001. From a Pemed listing.
Sensormedics VMax V62J Autobox Plethysmograph, 2003. Found at a Pemed.com listing.
Jaeger Masterscreen Plethysmograph, circa 2005. From a DotMed listing.
PFT History by Richard Johnston is licensed under a Creative Commons Attribution-NonCommercial 4.0 International License.